Will a magnet at the oil level be able to pick up ferrous wear particles?
火博体育下载Yes. Even though ferrous particles are more likely to sink than float, testing of the CMP has proven that the effects of both oil viscosity and turbulence will suspend wear particles so that they can be easily captured and viewed by the CMP’s magnetic plug.
What machine port hole diameters can be used with the CMP?
The CMP is currently designed for the following port sizes:
You may use additional bushings to expand the thread diameter but not火博体育下载 to reduce it. Reducing the diameter will cause errors in oil level measurement.
What if I need additional CMP parts?
If you need additional parts not shown online please contact us directly and we will be happy to assist you.
Why would I need a Copper Corrosion Indicator?
A corrosive condition can exist due to one or more of the following:
- Corrosion inhibitors become depleted (normal aging process of the oil)
- The base oil degrades by oxidation producing corrosives organic acids
- Corrosive agents are ingested from the environment (acids, water, salt, bacteria, etc.)
Corrosion can result in chemical damage to machine surfaces including cupric metals like copper, brass and bronze. When this happens there is no easy way to determine the onset of corrosion within your machine, risking catastrophic failure. The Copper Corrosion Indicator will tarnish when these abhorrent conditions exist, providing an early indication of a corrosion. This enables corrective action to be taken before irreparable damage occurs to the machine.
How does the CMP detect varnish?
The CMP comes with a Steel Corrosion Indicator (a Copper Corrosion Indicator is also available). If varnish is observed on the Corrosion Indicator through the Sight Glass, this is an indicator of a current or potential varnish problem. Varnish on the Corrosion Indicator = Varnish on the Inside of your Machine.
Is the oil in the CMP representative of the oil within the machine?
火博体育下载The CMP Sight Glass provides a visual inspection of the oil at the oil level. Oil level observations are an excellent way to inspect for foaming, discoloration and wear particles. A Sample Valve with a pilot tube is also provided with the CMP. This Pilot Tube can be placed deep within your machine below the oil level for live zone sampling.
How often do I need to clean the sight glass bowl?
The CMP should be cleaned whenever the Magnetic Plug and Corrosion Indicator are unable to be seen through the Sight Glass when rotated out of the oil.
Can the CMP be used with a pressurized system?
火博体育下载Pressure testing the CMP is currently underway.
Is the CMP compatible with synthetic lubricants such as polyalphaolefins (PAO) and phosphate esters?
火博体育下载The materials used in the CMP are compatible with mineral oils and most synthetic lubricants (e.g., PAO). Certain synthetics, such as phosphate esters, may exhibit marginal incompatibility. When in doubt it is recommended that compatibility be confirmed by contacting the lubricant supplier or performing compatibility tests in the laboratory.
Will the CMP leak if the oil sump is overfilled causing the sight glass to be completely filled with oil?
In normal use the CMP will not leak if filled above the oil level, however it is not recommended that this be a standard practice. This is also dependent on the tightness of the CMP threads and use of thread sealer.
How far below the oil level should the tip of the pilot tube be located?
The Pilot Tube tip should never rise above the oil level during standard operating conditions. If this level is not known then it is best to bend the Pilot Tube as far down as possible without interfering with internal machine components or hindering installation.
Can the CMP be exposed to direct sunlight?
火博体育下载Yes, the CMP can be exposed to direct sunlight.
Can the CMP be exposed to high pressure washing?
火博体育下载The CMP can be exposed to high pressure washing but it is not advised to spray the CMP directly with wash fluids at close range.
Can the CMP be exposed to common detergents and other chemicals mixed with water in high pressure washing?
火博体育下载Compatibility of the acrylic sight glass and elastomers used in the CMP should be confirmed.
Will the CMP sight glass melt if the oil gets very hot?
The CMP can withstand extreme heat. It would take temperatures exceeding 320⁰F for the Sight Glass to begin melting.
Can I install the CMP without lowering the oil level in my sump first?
Yes it is possible but not advised due to the amount of oil that might be lost and the associated hazard of leakage.
How often should I inspect oil conditions through the CMP?
We advise daily inspections of the CMP for early detection of problems with the oil and machine.
Won’t the threaded ports on the Sight Glass be potential contaminant entry points?
We estimate that at least ninety-five percent of the inspections of the magnet and corrosion indicators can be done without exposing the oil to the atmosphere (e.g., visually through the sight glass). The only time the magnet or indicator needs to be removed is when a real issue has been observed (as viewed through the sealed sight glass). If such an issue exists, either related to a problem with the oil (e.g., varnish or loss of corrosion suppression) or the machine (e.g., abnormal wear), we believe that early detection commands a far higher premium than a minor and temporary exposure of the oil to atmospheric contaminants.